Download HP A RF IMPEDANCE-ANALYZER OPERATION SM service manual & repair info for electronics experts. The HP A RF Impedance Analyzer measures 14 parameters Parameter measured: IZI, Swept measurement: manual or automatic sweep from start to. Agilent A Test Fixture. Operation and Service Manual. Third Edition. Agilent Part No. February Printed in Japan.
|Published (Last):||19 September 2008|
|PDF File Size:||2.98 Mb|
|ePub File Size:||8.74 Mb|
|Price:||Free* [*Free Regsitration Required]|
Most Active Software Boards: Good luck with it Accepted by topic author Matt C.
Agilent / HP 4191A RF Impedance Analyzer
Sorry, that last line of code there is supposed to read: The forums emoticons took over for a bit there. IIRC, the sweep parameters were something like this: You really need the manual.
Matt – you’re right, the commands for the A are completely different from the – probably because of its age. I have not worked mznual that instrument, but the Agilent web site has the operating and service manual here.
Would you mind posting the code you used on the A? In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows: Manua suspect that this difference might be causing my problems. Message 3 of 6.
Keysight / Agilent A – In Stock, We Buy Sell Repair, Price Quote
In the hp41911a above, MHz is an arbitrary frequency. Message 4 of 6. Has anyone done anything on interfacing with the A since Matt C’s last post? However, rather than being returned the data at MHz, I am returned the data at whatever frequency the analyzer is currently set at.
I am able to communicate with the device through the GPIB interface, and am also able to retrieve information.
Thanks for your help guys. Message 5 of 6.
I have worked with the A and, as I recall, the commands are nothing like any other instrument I have seen. I think the would probably take the old commands, but the ALSO has new commands.
Message 6 of 6. Should I be looking at doing it with LabView or something else? It’s 84MB, so I hope you have a fast link. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
Guess you never know until you try! If not, up4191a would creating a driver for it entail? Message 2 of 6. When I gave it a try, though, it worked fine. Message 1 mahual 6. In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows:.
I went through the manual before, but when I saw manjal GPIB section, the strings were so convoluted and so unlike anything I’ve ever seen before that I assumed it was referring to something different.